I’ve been meaning to write a follow up to my earlier post about the Feature Tech antenna analyser. The short summary of that post was that when I hooked up the antenna analyser to a mobile whip on a magnetic base, I saw all sorts of SWR values which were often not repeatable and changed even as I was taking my hand away from the meter. I was disappointed with the device.
Fast forward a few weeks, in a QSO with Martin VK6ZMS and he was strongly of the opinion that I was doing myself a disservice by running a mag mount without grounding the antenna to my vehicle close to the base of the antenna. Another clue was one idle afternoon at home, I threw the analyser on my WARG Pogo Stick 2m antenna and it had none of the problems I had seen. The SWR readings were reliable (you could read of a value for a frequency, change the frequency then change it back and get the same SWR value) and did not alter as I handled the coax or moved away from the meter. One of the features of this antenna is that it has an coax choke at the base.